Pliers Electrical 43313.1 (433131, 43313-1, 43313 1, 43313,1)
Pliers Electrical 43313.1 designed to measure the voltage on the underground metal structures relative electrode (building construction) and its components, voltage and AC current, RMS voltage and AC sinusoidal curve, DC resistance.
On measuring ranges capacity building provided the possibility of introducing a compensating voltage.
Provides the ability to polarization of the auxiliary electrode and measuring the voltage on it with respect to the reference electrode, that is, the voltage on the construction with the exception of resistive component (polarization potential).
The main field of use of the instrument Clamp Electrical 43313.1 - in specialized units of electrochemical protection in assessing the extent of corrosion, protection from her metal structures (pipes, cables, tanks, vessels, containers, etc.) and maintenance of protective equipment in the laboratory, field and production conditions.
Electrical appliance 43313.1 pliers can be used as a standard meter for the operation and maintenance of electric and radio.
Technical characteristics of the instrument Clamp Electrical 43313.1:
- DC power (class) - 0.2 mA 2 mA, 2 mA to 20 mA, 20 mA, 2000mA, 2000mA-20000mA (1.5 / 0.5);
- AC power (class) - 0.2 mA 2 mA, 2 mA to 20 mA, 20 mA, 2000mA, 2000mA-20000mA (2.0 / 1.0);
- DC (class) - 0.2 V-2B, 2B-20V, 20V-200V (0.4 / 0.2);
- AC voltage (class) - 0.2 V-2B, 2B-20V, 20V-600V (2.0 / 1.0);
- DC resistance (class) - 0.02 kW, 0, 2 kW, 0.2 kW-2K, 2 k, 20 k, 20 k, 200 k, 200 k-2000 kW, 2000 kW, 20,000 kW ( 0.5 / 0.25);
Potential construction - ± 2V-± 20V;
Polarization potential of the instrument Clamp Electrical 43313.1 - ± 2V;
Input DC resistance - 10 MW;
Input DC resistance in the range of the instrument Clamp Electrical 2B 43313.1 - more than 100 MW;
Food - independent;
Dimensions of the unit ticks Electrical 43313.1 - 250h120h125mm;
Weight - 1.7 kg;
The manufacturer guarantees that the quality of the instrument Clamp Electrical 43313.1 technical requirements specifications provided the customer terms and conditions of storage, transportation, installation, operation, and its specified operational documentation.