Cymometer narrow profile TS1736 (TS-1736, TS 1736)
Suitable for frequency measurement in AC networks, and also for signaling the measured values from the output of the specified values.
Measuring ranges: 45Hz to 55Hz, 55Hz, 65Hz or 350Hz-450Hz.
Rated voltage: 127V, 220V or 380V.
Accuracy class 0.5.
Cymometer narrow profile TS1736, shield, shock-resistant, vibration-and shock-resistant, Rectifier, with light indicator, designed to measure the frequency of AC networks, as well as for signaling on the measured value of the specified field values.
Accuracy class - 0.5. Frequency can work complete with a regulating device whose input is consistent with the parameters of photoresists.
Cymometer narrow profile TS1736 designed to operate in ambient temperatures from minus 30 to plus 50 ° C and relative humidity up to 100%.
Cymometer narrow profile TS1736, produced in tropical version, designed for operation at ambient temperatures from minus 10 to plus 55 ° C and relative humidity up to 100% at 35 ° C and is designated TS1736 T2. Designation of frequency, additional devices and design features are shown in Table. 1.
Measuring ranges of the frequency TS1736: 45Hz to 55Hz, 55Hz, 65Hz or 350Hz-450Hz. Nominal voltage controlled network: 127V, 220V or 380V.
Maximum permissible error of frequency TS1736 is ± 0,5%.
The limit of permissible error of the contact device to the frequency of modifications, CP and TC is ± 1,0%,
For Tolerance In photocontact devices receive the difference between the actual value of the measured values at the time of operation and the amount by which the pointer is set contact device (edge filter).
The limit of additional error frequency TS1736 caused by:
changes in ambient temperature of (20 ± 5) ° C and any temperature within the range of from minus 30 to plus 50 ° C at every 10 ° C change in temperature equal to ± 0,25%;
voltage deviation of ± 10% of nominal, equal to ± 0,5%;
the influence of an external magnetic field with induction 0.5 mT is ± 1%;
frequency deviation of 45 ° from the normal position in either direction is ± 0,5%.
Primary and secondary errors are expressed as a percentage of the final value of the measuring range.
Response time is less than 3s. Warm-up time 15 min. Dielectric strength 2kV. The length of the scale corresponding to the measurement range, 118mm.
The light source used incandescent 0P6-3-B5.
Meals Illuminator from the transformer P1710 (10 bulbs), P1711 (20 bulbs), P1712 (50 bulbs) or from any DC or AC voltage of 6 V. Mains voltage should be from plus 10 to minus 15% of nominal.
The output current of the contact device to device TS1736 modifications, CP and TC changes to 30MKA (not more) in the dark state and to 250mkA (at least) in a light state when powered from the mains voltage of 12V. Areas of possible displacement of filters are shown in Table 2. Overall dimensions: indicating device 160h30h272, 5; additional devices 110h155h95. Weight, max: 1.5 kg indicating device; additional device 1,7 kg.