High frequency inductance box R594 (P 594, P-594, p 594, p-594, p594)
Designed for use as a model unit or element of AC measuring schemes with frequency from 10 kHz to 1000 kHz.
Basic accuracy error ± 1%
High frequency inductance box R594 with ferromagnetic cores is designed for use as a model unit or element of AC measuring schemes with frequency from 10 kHz to 1000 kHz.
Maximum inductance value 1111mkH.
Quantity of decades - 4: 100mkH-1000mkH; 10mkH-100mkH; 1mkH -10mkH; 0.1 mkH- 1mkH.
Decades acceptable basic error limit 100mkH -1000mkH; 10mkH-100mkH; 1mkH-10mkH is ± 1% of inductance nominal value.
Variometers acceptable basic error limit is ± 0.05 mkH.
Normal frequency 10 kHz.
Operating frequency range:
10-100 kHz - for decade 100-1000 mkH;
10-300 kHz - for decade 10-100 mkH;
10-1000 kHz - for decade 1-10 and 0.1- 1mkH.
Maximum operating voltage 250 V.
Maximum current 500 mA.
Box’s quality (Q) is at least: 80 at frequency 100 kHz for maximum decades set 100-1000mkH and 50 at frequency 300 kHz for maximum decades set 10-100 mkH.
Dimensions do not exceed 500x135x265 mm.
Weight does not exceed 8 kg.
Photos P594 (Р594)