Oscilloscope S1-127/1 (S1 127/1, c1-127/1)
To monitor and measure electrical signals scale of 4 mV to 300 V and a duration of 20ns to 2s in the frequency range from 0 to 50MHz.
Screen Size: 80 mm (10 cases) x 60mm (8 cases)
Coefficient of variation: from 1 mV / div to 5 V / div
Sweep: from 0,05 μs / div to 0,2 s / Affairs
1.1. Oscilloscope S1-127/1 is used to monitor and measure electrical signals from the scale 4 mV to 300 V and a duration of 20 ns to 2 in the frequency band s 0 to 50 MHz.
1.2. Service conditions use: for oscilloscope S1-137
Increased ambient temperature, ° to 50;
The reduced air temperature, ° to minus 30, high moisture, 98% at 25 ° C;
Pressure, mm Hg 400 to 780
Oscilloscope S1-127/1 high ambient temperature, ° to 40;
Low temperature of the ambient air, ° C plus 5;
Humidity, 80% at 25 ° C;
Barometric pressure, mm hg from 400 to 780.
1.3. Oscilloscope is used in the production, development and operation of electronic equipment.
2. Technical data, oscilloscope S1-127/1
2.1. Working part CRT oscilloscope horizontal 80mm (10 cases);
Vertically 60mm (8 cases).
Note. Sizes are in parentheses display into bars.
2.2. Linewidth beam no more than:
0.8 mm with a coefficient of variation of 0.005 - 5 V / div;
0.2 Affairs at Coefficient of variation 1 and 2 mV / div;
geometric distortion not more than 3%.
2.3. Minimum frequency of the sweep, which provides a monitoring of the signal at a rate of sweep 5 ns / cases up to 1 kHz with an artificial darkening of the screen.
2.4. Coefficient of variation of channels A and B are set twelve calibrated steps from 1 mV / div to 5 V / div and gradually increase relative to the calibration values not less "than 2.5 times.
2.5. Limit of acceptable values of the coefficients of variations of the basic error of 0.005 - 5 V / div is + -3%, deflection factors 1 and 2 mV / div ± 4%;
Limit of acceptable deviation of error coefficients of 0.005 - 5 V / div in the working conditions of use is
+ -4.5% Deviation coefficients 1, 2 mV / - + -6, 1:10 divider for applying said error is not increased by more than 1%.
2.6. The value of the synchronous hum no more 0.2 Affairs + 1 mV.
In the working conditions of application release, the unevenness in the area of establishing not standardized, settling time less than 50 ns, non-uniformity - 3%.
2.7. The decline at the top of nx gated entrance in the time interval No 0,5 ms more than 5%.
2.8. Displacement of the beam of each channel on the CRT screen is not more than:
1 cases due to the input current, and switching the switch "V / div";
2 cases with inverting signal channel B;
2.9. Limits vertical displacement of the beam in each channel about the middle of the working of CRT not less than 60 mm.
2.10. Input parameters for each channel vertical deflection:
With direct input Input resistance is equal to (1 + 0.03) mW, the input capacity of not more than - 25 p f;
With 1:10 divider input resistance is equal to (10 + -0.3) and the IOM, the input capacitance - no more than 15 p f.
2.11. Permissible total value of the DC and AC voltages from direct input of each channel vertical deflection should not exceed 50 V, with a divider 1:10 - 300 V.
2.12. Factor isolation between channels, the vertical deflection of at least 50 at a frequency of 15 MHz and at least 25 at a frequency of 50 MHz.
2.13. Apparent time delay signal on the screen of the CRT the sweep to the level of 0.1 is not less than 15 ns.
2.14. Tract of vertical deflection provides the following modes Hours:
Monitoring a channel signal; monitoring signal channel b;
The algebraic summation of the signals of channels A and B, alternately or broken circuit switching and and b;
Changing the polarity of the signal in channel B. 2. 15. Tract provides a self-oscillating horizontal deflection, Standby mode and single work.
2.16. Sweep set twenty-one calibrated steps of 0,05 μs / div to 0,2 s / cases. Envisaged gradual increase sweep no less than 2.5 times the calibrated position in the oscilloscope is tenfold stretching sweep.
2.17. Limits of acceptable basic error of the sweep are + -3% with no stretch marks and x with p astyazhkoy.
The limits of acceptable error sweep in the working conditions of use are + - 4.5% without stretching and + -6% with a banner.
The limits of acceptable basic error of measurement of time intervals to sweep 0,05; 0,1; 0,2 μs / cases enabled stretching are + -5%, in the working conditions of use + -7.5%.
2.18. Moving beam provides horizontal alignment start and end sweep line with the center of the screen.
2.19. Coefficient of variation of the horizontal sweep mode by external signal input " "No more than 2 V / div, bandwidth tract horizontal deflection in sweep mode from an external signal 20 Hz 3 MHz.
2.20. Parameters Inputs X and Z respectively: input resistance (1 + 0.1) Megohm (100 + -20) com, input capacitance not more than 70 pF.
1 cases, gradually varying coefficients of variation in the range of 0.005 - 5 V / div.
When the coefficients of 1, 2 mV / div offset is not specified.
2.21. Tract horizontal deflection provides the following types of synchronization of the sweep:
External synchronization, the synchronization of channel and, synchronization of Channel B, synchronization of network.
2.22. Frequency synchronization of 10 Hz to 75 MHz. In self-oscillating mode synchronization is done with Signals frequency of at least 50 Hz.
2.23. Emission limits synchronization
When the internal clock minimum level of no more than 0.8 cases, the maximum - at least 8 cases;
When the external clock minimum level does not exceed 0.2 V, the maximum - not At least five
The instability of the synchronization is not more than 0.2 cases of the scale of the CRT screen.
2.24. Oscilloscope Calibrator provides a square wave at the output of "square wave" with a repetition rate of 1 kHz and 0.6 V. ampdlitudoy
The limits of permissible values of the basic errors of pulse repetition rate and amplitude of the calibrator are + -1%, the limits of permissible error values of the amplitude and frequency of the pulse calibrator in the working conditions of application are + -1.5%.
2.25. Electrical isolation between the input supply chain network connector and the body of the oscilloscope can withstand without breakdown and surface flashover test voltage On current frequency of 50 Hz, the effective value of 1.5 kV and 900 normal to the moisture.
2.26. Electrical insulation resistance of the oscilloscope power relative to the body not less than:
under normal weather conditions - 20 MW;
at elevated ambient temperature - 5 MW;
in humid ambient air - 2 IOM;
2.27. Oscilloscope provides its specifications within the norms established that, after the time set the mode of 15 min.
2.28. Oscilloscope allows continuous operation in the working conditions for a time at least 16 h while maintaining its specifications within the standards set specifications.
Note. Continuous operation excludes yourself time to establish the operating mode of the oscilloscope.
2.29. Oscilloscope retains its specifications within the standards set specifications, when it feeds on AC power:
Voltage (220 ± 22) V, (50 + 1) GCI harmonic content up to 5%;
Voltage (220 + -11) V, (115 + -6), frequency (400 ± 10) Hz and harmonic content to 5%;
2.30. Power input of the oscilloscope network Power at rated voltage is 50 V. A network of re alternating voltage of 220 V, 50 Hz and 30 W, and the DC voltage of 27 V (for oscilloscope S1-127).
2.31. Oscilloscope retains its specifications within the standards set by the one after replacing the CRT in it.
Allowed podregulirovanie oscilloscope using the available enforcement adjustment provided for the electrical circuit diagram and instructions for use of the oscilloscope,
2.32. Modes of vacuum, semiconductor, electric components meet the standards established in the standards and that they are provided, and in the working conditions for a time of continuous operation, set forth in Item 2.2V.
2.33. Voltage industrial radio interference oscilloscope in decibels:
at frequency 6 100 MHz u = 60, where F - Frequency, MHz.
2.34. MTBF oscilloscope T of not less than 7000 h.
2.35. Gamma-percent life of at least 10,000 with γ = 90%.
2.36. Gamma-percent service life of 15 years at γ = 95%.
2.37. Gamma-term interest rate persistence of at least 10 years for heated storage and 5 years for unheated storage at γ = 99%.
Oscilloscope without silica and covers long-term storage not to be.
2.38. Average recovery time oscilloscope over 3 h.
2.39. The probability of no latent failures of at least 0.92 for the calibration interval 24 months, with an average utilization rate of Ki = 0.04.
2.40. Overall dimensions, mm, not more:
Oscilloscope - 295h139h495;
Personnel Package - 585h185h595;
Shipping container - 815h349h670.
2.41. Weight Oscilloscope not more 6 kg;
Weight oscilloscope with service pack no more than 16 kg;
Weight oscilloscope transport container no more than 40 kg.