Oscilloscope S1-101 (S1 101, S1101, c1-101, c1 101, c1101)
Suitable formulas for the study of periodic electrical signals by visual observation and measurement of the amplitude in the range from 0.01 V to 300 V, and time intervals of 0.3 · 10 -6 s to 0.4 s.
Frequency range from 0 to 5 MHz.
The range of values of the coefficient of variation: 0.005, 0.01, 0.02, 0.1, 0.2, 0.5, 1, 2, 5 V / div.
Basic error limit of deflection factors should be ± 7%.
The margin of error with a remote 1:10 divider must be ± 7%.
Yield bias error coefficients in the working conditions must be ± 8%.
Oscilloscope S1-101 universal chip is designed to study formulas periodic electrical signals by means of visual observation and measurement of the amplitude in the range from 0.01 V to 300 V, and time intervals of 0.3 · 10 -6 s to 0.4 s, the frequency range from 0 up to 5 MHz
According to signal fidelity, measurement time and amplitude values oscilloscope S1-101 belongs to the class III GOST 22737-77 "cathode-ray oscilloscope. Nomenclature parameters. General requirements."
Operating conditions n ribora Oscilloscope S1-101
Operating ambient temperature from -30 ° C to +50 ° C with power supply I22.087 457 - from -20 ° C to +50 ° C, relative humidity up to 98% at temperatures up to +35 ° C, with the block Power I22. 087,457 - up to 80% at +35 ° C device operates normally after exposure (in the packing box) shock
repeated action with acceleration up to 147 m / s 2 and a pulse duration of 5 ms and 10 ms, a single action with acceleration up to 735m / s 2 and the duration of 1 ms to 10 ms.
Oscilloscope S1-101 resistant to cyclical changes in ambient temperature from -50 ° C to +65 ° C with power supply I22 087,457 - from -50 ° C to +60 ° C.
Oscilloscope S1-101 can be used in the design, configuration and adjustment of electronic circuits, for inspection and repair of instrumentation and automation of various devices in the laboratory and in the field, especially in hard to reach places when setting up and testing of computing devices
Range rejection coefficient. 0.005, 0.01, 0.02, 0.1, 0.2, 0.5, 1, 2, 5V/div.
Basic error limit of deflection factors should be ± 5% margin of error with external divider must be January 10 ± 7% margin of error coefficients of variation in working conditions must be ± 7,5%
In all positions of the switch "V / DIV" permitted the rise or decline the leading edge of the rectangular pulse of ± 5%, with external divider 1:10 - in the range of ± 20%
Rise time of the channel vertical deviation must be less than 70 ns with the direct input and no more than 100 ns with a divider 1 x 10
Release of the transient response of the channel vertical deflection should be no more than 5% - in all positions of the switch "V / DIV", 8% - with external divider 1 10.
Time of the transient characteristics of the channel vertical deviation should be less than 210 ns, with external divider 1 × 10 - less than 250 ns.
Irregularity of the transient response to be no more than ± 3%
Recession vertices (with closed inlet) must be no more than 10% at a test pulse duration 10 ms
Parameters input channel vertical deflection:
Input resistance at the open inlet (1 ± 0,02) megohms,
input capacitance (40 ± 4) pF
External divider must have an active input impedance (1 ± 0,03) MW and the input capacitance of less than 15 pF.
Permissible total value of the DC and AC voltage gated entrance ("~") channel oscilloscope vertical deflection device S1-101 should not be more than 200, and with a divider 1 · 10 to 300 V.
The limits of the vertical displacement of the beam shall be not less than two times the rated vertical deflection
The range of values of the coefficients sweep
0.1, 0.2, 0.5, 1, 2, 5, 10, 20, 50 ms / div,
0.1, 0.2, 0.5, 1, 2, 5, 10, 20, 50 ms / div
0.1, 0.2 s / div.
Basic error limit of sweep must be ± 3% at the measured image size from 6 to 7 bar working part of the sweep, and ± 4% for all other image sizes from 4 to 8 bars in the range of 2 ms / div - 50 ms / div ± 5% the range of 0.1 - 1 mkc / div
Sweep 0.1 s / div and 0.2 s / div are screening. The margin of error in scanning the coefficients operating conditions must be ± 4,5% measured with the image size of 6 to 7 bar and the working part of the sweep ± 6% for the remaining image sizes of from 4 to 8 bar in the range 2 ms mks/del-50 / Affairs, ± 7,5% in the range of 0.1 - 1 ms / div
Margin of the beam in horizontal alignment should provide rocked and the end of the working parts of the scanning center of the screen. Parameters of the internal clock synchronization frequency range should be from 20 Hz to 5 × 10 6 Hz, the minimum and maximum timing should be 3 mm (06 cases) and 30 mm (6 cases), respectively, the instability of the synchronization should be no more than 1 mm (0 , 2 cases). External synchronization parameters: frequency range of external synchronization must be 20 Hz to 5 × 10 6 Hz, the minimum and maximum timing should be 0.5 V and 20 V respectively, the instability of the synchronization should be no more than 1 mm (0.2 cases) External trigger input for the input "Ext January 1", the input resistance - not less than 50 ohm input capacity - not more than 30 pF for input "Ext January 10," Input resistance - not less than 750 ohms, the input capacitance - more than 20 pF.
Effective screen of the oscilloscope should be 40mm at 8 bar (price for 1 division - 5 mm) horizontally, 30 mm or 6 bars (dividing the price of 1 - 5 mm) vertically. The line width of the beam shall be not more than 0.6 mm.
Short term drift Oscilloscopes S1-101 after 5 min of heating must be not more than 1 mV for 1 min operation. Long-term drift - 5 mV / hr for 1 hour
The displacement of the beam line at the transition from one value to another coefficient of variation should be no more than 1 div
The displacement of the beam line due to the input current does not exceed 1 cases.
The displacement of the beam line when the supply voltage must be no more than 0.2 cases.
Periodic and (or) random deviations should be no more than 3% of the rated deflection.
Adjust the brightness to change the brightness of the image to ensure the total absence to a comfortable to watch.
The internal power supply shall generate a calibrated rectangular pulses with a repetition rate of 1 kHz and an amplitude of 0.05 V and 1 V
The margin of error pulse amplitude calibrator should be
± 1,5% - in normal circumstances,
± 2% - in working conditions.
The margin of error pulse frequency calibrator should be
± 1% - in normal circumstances,
± 1,5% - in working conditions.
The maximum amplitude of the signal should be no more than 30 - to the vertical position of the inlet channel, 300 V - input divider 1 10.
The amplitude of the sinusoidal voltage must not exceed 15V and 150V respectively.
The amplitude of the voltage sweep, extracted at the nest "OL" should be at least 2 V at no less than 20 ohms with an output capacity of no more than 20 pF. Overall dimensions (281 h155x 69) mm. Overall dimensions in the packing box - (526 x265h200) mm. Overall dimensions of shipping containers - (725X406X 323) mm. Weight of the device should be less than 1.8 kg, with power supply I22.087.459 - not more than 2,3 kg, with power supply I22 087,457 - less than 2,5 kg, with a divider I22.727.095 - not more than 1.9 kg.
Weight of the device Oscilloscope S1-101 in the packing box should be no more than 10 kg.
Weight of the device in transport containers should be no more than 22 kg.
The power consumed by the device from AC mains at the rated voltage should be no more than 18VA.
Current consumed from the DC power at a voltage of 12 V and 27 V must be no more than 0.70 A.
Oscilloscope S1-101 should retain its specifications within the standards set by the technical specifications for their nutrition from AC with a frequency of (50 ± 0,5) Hz voltage (220 ± 22) and harmonic content up to 5% or frequency ( 400 ± 10) Hz and voltage (115 ± 5,75) B and (220 ± 11) and a harmonic content up to 5% of the constant current sources (12 ± 1,2) B and (27 ± 2,7) V from the power supply I22 087,457.
Note: When you export supply and availability requirements in the purchase order, it is possible revision of the device to ensure its supply of AC frequency (50 ± 0,5) Hz (230 ± 23) V or (240 ± 24) and the content harmonics up to 5%. These devices are not intended for inclusion in the AC line voltage of 220 V and 115 V.
Oscilloscope S1-101 should be capable of continuous operation in the working conditions for a time at least 16 hours while maintaining its specifications within the standards set specifications.
While allowing the normal modes of vacuum, semiconductor, electric components within the norms, standards and technical specifications for these
When the instrument oscilloscope S1-101 unit I22 087 457 of length should be in normal conditions for at least 1 h at a temperature of +50 ° C for at least 40 minutes at -20 ° C for at least 20 minutes.
Mean time to failure of the device (It) should be no less than 3500 hours The device should be capable of long-term storage in a heated and unheated storage capital.
Shelf life Oscilloscopes S1-101 in a heated storage for at least 12 years of age. Shelf life major appliance in an unheated storage for at least 10 years of age.
Shelf life with the power supply unit I22 087 457 at least 3 years Average lifetime without power supply 087,457 I22 at least 10 years of age. Average life 10,000 hour average life of the appliance to the power supply I22 087,457, including storage, 3 years. Within 3 years of power supply I22 087,457 to withstand at least 150 cycles (charge-discharge).