Measuring the parameters of electron tubes L3-3 (L33, L3 3)
Measuring the parameters of electron tubes L3-3 designed to measure the main electrical parameters of vacuum tubes, as well as to remove the static characteristics.
Measuring the parameters of electron tubes L3-3 allows measurement of low-power (with power dissipation up to 25W at the anode) of vacuum tubes in the modes or modes of specifications or state standard.
Measuring the parameters of electron tubes L3-3 can be used na warehouses and bases consumer electronic tubes, repair workshops, laboratories, as well as companies that develop and produce electronic equipment.
Measuring the parameters of electron tubes L3-3 can be used in climatic conditions: ambient temperature from -10 to +40 ° C, relative humidity of (65 ± 15%) and atmospheric pressure (750 ± 30) mm Hg. Art.
Note. If the equipment is brought into a room with an ambient temperature above 0 ° C from the street or from another room, where the air temperature is below 0 ° C, the meter before use is necessary to keep in the room for at least 2 days.
Measuring the parameters of electron tubes L3-3 is resistant to vibration during transport including 80-120 beats per minute and acceleration of 30 m / s 2, and the vibration frequency of 30 Hz and an amplitude of 0.4 mm.
Meals meter is powered by the AC voltage (127 ± 12,7) V (220 ± 22) V (50 ± ± 0,5) Hz or AC voltage (115 ± ± 5,75) and frequency ( 400 ± 12) V
Measuring the parameters of electron tubes L3-3 has a notional scale with 75 divisions, which corresponds to the following (standard value:
when measuring the anode voltage Ua and voltage stabilization U stub. (Potential ignition Uzg) Zener - 15V, 75V, 150V, 300V;
when measuring a second voltage grid Ug 2 - 75V, 150V, 300V, Voltage measurement grid of the first Ug 1 - 1.5 V, 3.0 V, 7.5 V, 15.0 V, 30.0 V, 75.0 V;
when measuring the filament voltage Uh, Uf - 3,0 V, 7.5 V, 15.0 V when measuring the anode current Ia and the emission current Ie diode - 1.5 mA;
3.0 mA, 7.5 mA 15.0 mA 30.0 mA 75.0 mA 150.0 mA;
when measuring the grid current Ig second 2 - 0.75 mA, 1.50 mA, 3.00 mA, 7.50 mA, 15 mA;
when measuring the rectified current I sn - 150mA; 300mA; when measuring supply voltage circuit (250V) - 300V.
reverse current measurement I g first grid 1 and the anode current I characteristic at the beginning of a0 - 0.75 mA, 3.00 mA; 15.00 uA; 30.00 uA; uA 150.00;
when measuring the leakage current between the electrodes Ileak -150 uA;
when measuring the slope of the anode-grid characteristic S (slope) - 0.75 mA / V, 1.50 mA / V, 3.00 mA / V, 7.50 mA / V, 15.00 mA / V, 30.00 mA / V 75, 00mA / V.
Limit of acceptable basic error of measurement:
Device control reverse current grid of the first, at the beginning of the anode current characteristics and leakage current between the electrodes - no more than ± 2,5% of full scale of each scale;
slope control devices - no more than ± 2,5% of full scale of each scale.
Normal conditions are the following: the ambient temperature (20 ± 5) ° C, relative humidity of (65 ± 15)%, atmospheric pressure (750 ± 30) mm. Hg. Art.
Limit of acceptable additional error of measurement:
Devices kontp Olya constant voltage mode, the anode current control devices, emission current grid current and the second rectified current rectifier tubes in the temperature of the ambient air (-10 to +40 ° C at a relative humidity of (65 ± 15)% should be no more than ± 1,2% of full scale for every 10 ° C change in temperature.
The limit of acceptable measurement error additional control devices (reverse current steepness first grid, the anode current at the beginning of the characteristics and leakage current between the electrodes within the ambient temperature from -10 ° C to +40 ° C at a relative humidity of (65 ± 15)% should be no more than ± 2% of full scale per 10 ° Sizmeneniya temperature.
The AC power circuits izmeritelyaparametrov electronic tubes L3-3 (secondary voltage) via a switch NETWORK (held constant, which corresponds to position the arrow on the meter gauge of the scale with an accuracy of ± 1 division by holding down the network. At rated voltage mains switch must NETWORK be in position 7.
Measuring the parameters of electron tubes L3-3 supplies the electrodes (audited lamps following voltages:
on the heat:
permanent - from 1V to 14V at load current up to 1.2 A; variables - 2.5 V, 3.0 V, 4.5 V, 5.5 V, 7.0 V at load current up to 3A;
10V, 13V load current up to 1.7 A, 15V, 25V load current up to 1.3 A;
first on the grid:
permanent - from minus 0.5 to minus minus 65V and 100V, the second on the grid:
permanent - from 10V to 300V with a load current up to 15 mA;
permanent - from 5V to 25V at load current up to 50 mA;
from 25V to 300V with a load current up to 100 mA;
variables - 2 to 350V, 400V for 2, 2 for 500V.
To ensure self-bias on verifiable lamp in the meter has the following values of the cathode resistors 30, 50, 68, 75, 80, 100, 120, 150, 160, 200, 220, 400, 500, 600, 600 ohms.
Measuring the parameters of electron tubes L3-3 provides a measurement of electron tubes in accordance with the attached list the following:
for diodes - emission current or anode current;
in transistors, double triode, tetrode, penthodes and combined lamps - current anode current of the second grid, the reverse current of the first grid, the slope of the anode-grid characteristics, slope characteristics of the heterodyne frequency converter tubes, the anode current characteristics at the beginning or the blocking voltage grid of the first ;
with Zener - potential ignition voltage stabilization, stabilization of voltage changes when you change the value of the load current;
in rectifier tubes - rectified current.
In addition, measuring the parameters of electron tubes L3-3 can be measured in heated-lights: the leakage current between the electrodes at voltages of 100V and 250V (when checking the leakage current between the cathode and heater-plus on the cathode minus the heater on).
The meter allows you to remove the static characteristics of the lamps. The maximum power consumption meter at nominal line voltage does not exceed 300VA checking of all types of lamps, except the lamp 5TS3S when checking lamp 5TS3S - 450VA.
Measuring the parameters of electron tubes L3-3 is designed for continuous 8 hours each, not including the time samoprogreva when testing different types of lamps with an anode current of up to 100mA. Allowed. 2-Hour Meter for continuous testing of lamps of the same type with an anode current of 100 mA to 150 mA. Samoprogreva time is 30 minutes. A break to re-enable - not less than 1 hour
Measuring the parameters of electron tubes L3-3 maintains its electrical parameters after changing the entire set of vacuum tubes with the appropriate regulators readjustment. The MTBF - not less than 3000 hours average life - at least 10 years of age. Average recovery time - no more than 60 minutes. Overall dimensions of no more than 515h320h230mm. Weight 20 kg.