Vibration measurement >> high-frequency accelerometers (AVS)
Made in: 2017.
Accelerometer high-frequency AVS 034-05
Also, this device can be called: AVS-034-05, AVS034-05, AVSO34-05, AVS 034 05, AVSO 34-05, AVSO-34-05, abc 034-05, abc-034-05, abc034-05.
AVS 034-05 high-frequency accelerometer piezoelectric is used for analysis of high vibration and shock acceleration.
The upper limit of the measured vibration accelerations - 50,000 m/s2.
The upper limit of the measured shock acceleration - 100 000 m/s 2.
Operating frequency range - 20 Hz to 30,000 Hz.
The sensing element - bending type.
Accelerometers AVS 034-05 performed by differential mechanical and electrical circuits.
High noise immunity.
Base - tight seal is made of titanium alloy.
Cable - antivibration.
Mount accelerometer AVS 034-05 - BONDED two screws with a ring or a special bracket into two seating surfaces.
The main element of the piezoelectric accelerometer is a drive of a piezoelectric material, which is used as artificially polarized ferroelectric ceramics. Subjected to a force (tensile, compressive or shear) piezoelectric material generates on their surfaces, which are attached to the electrodes, an electric charge proportional to the force exerted.
Akselerometr AVS 034-05 is designed to measure vibration accelerations on rocket and space technology, aircraft, machine under conditions of high electromagnetic interference.
The manufacturer guarantees compliance accelerometers AVS 034-05 all technical requirements if the consumer terms and conditions of use, storage and transportation, installed documentation manuals.