Dosimeter SINTEKS-DBG-01S (DBG01S, DBG 01S, DBG-01-S)
Dosimeter SINTEKS-DBG-01S is designed to measure the equivalent dose of external gamma radiation, display the result on the digital display and qualitative assessment of the intensity of gamma radiation with a sound alarm.
The dosimeter is intended for individual use to the rapid assessment of the radiological situation in the room or on the ground.
It also can be used in industrial (check scrap, berries and mushrooms, minerals) as well as for domestic purposes (checking food items, goods, transport, buildings).
Dosimeter "SINTEKS" is made of high impact polystyrene and is made in the form of a portable device that is easy to carry in your pocket.Only two body control make the unit convenient and affordable for people of all skill levels. Readings not only displayed on the scoreboard, but also accompanied by an alarm, which is useful in field conditions, such as mushroom pickers.In the forest, he will not need to retrieve the device every time out of his pocket - he warns of the danger itself.
The delivery of the dosimeter "SINTEKS" consists of three elements of power such as "Tom Thumb" and the book "Understanding the radioactivity to the public."
Technical characteristics of the devices dosimeters SINTEKS-DBG-01S:
Measurement range of equivalent doses of external gamma radiation - 0.10 mSv / h to 999mkZv / h;
Measurement range of equivalent doses of external gamma radiation is divided into three ranges:
- 0.01 mSv / h and 9.99 mSv / h;
- 0.1 mSv / h to 99.9 mSv / h;
- From 1mkZv / h to 999mkZv / h;
Limits of the relative error of the dosimeter device DBG-01S - no more than ± 25% at any point in the working range in the calibration of an exemplary source of the second class of cesium-137 in normal operation;
The limits of permissible additional error SINTEKS dosimeter device, due to changes in ambient temperatures ranging from -10 º C to +40 º C - ± 10% for every 10 º C;
The limits of permissible additional error caused by a change of the nominal supply voltage in the range 4.5 V ± 0,45 V - no more than ± 5%;
Energy dependence in the energy range - from 0.04 MeV to 1.3 MeV;
Operation mode setup time dosimeter SINTEKS device - no more than 1 minute;
Measurement time, depending on the selected range 3c-4c, 5c, 7c and 35c-37s;
Time of continuous operation with no evidence of instability than ± 5% - not less than 8:00;
Power - three recessed dry galvanic elements 316 total voltage 4.5 V ± 0,45 V;
Dimensions - no more 145h65h30mm;
Weight of the device dosimeter DBG-01S - not more than 0.25 kg;
With the release of the production of the dosimeter DBGB-01S exposed obligatory primary state verification, which is a relevant note in the manual.