Bridge AC R5016 (R-5016, R 5016, p5016, p-5016, p 5016)
For measuring the capacitance C and the loss tangent tg δ in parallel substitution inductance L and the loss tangent tg δ in series substitutional resistance R and the tangent of the phase shift tg φ.
Measurement ranges main parameters: C - from 10-14 F to 10-4 L - 10 -7 H to 10 2 GN; R - 10 -2 to 106 ohms.
Measuring ranges of related parameters: tgδ - from 10-4 to 1 and from 10-3 to 1; tgφ - from-3 to +10 ± 1.
Dimensions 490x210x495 mm, weight 35 kg.
Bridge AC R5016 designed to measure capacitance C and the loss tangent tg δ in parallel displacement, the Land inductance dissipation factor tg δ of substitution in series, resistance R and the tangent of the phase shift tg φ.
In terms of mechanical and climatic influences Bridge AC R5016 correspond to GOST 22261-76 (group 2).
Main technical characteristics
Measurement ranges main parameters: C - from 10 to 14 F to 10 F -4; L - 10 -7 H to 10 2 GN; R - 10 -2 to 10 ohms 6 ohms.
Measuring ranges of related parameters: tgδ - from 10 -4 to 1 and from 10 -3 to 1; tgφ - from -3 to +10 ± 1.
The number of decimal places of reference for C, L, R-five and tgδ and tgφ - four.
Bridge AC R5016 has three measurement modes: search mode, a mode of repeated measurements and tracking mode.
Starting the bridge manual, automatic and external. Selection of sub-band measurements automatically.
The device provides output measurements in BCD parallel-serial code 8-4-2-1.
Logic "1" corresponds to the voltage from 2.4 to 5.25 V, the logical "0" - from 0 to 0.4 V.
Pulse parameters, "Full-scale": the amplitude of 2.4 to 5.25, the duration from 4 to 10 ms, the rise time is not more than 0.5 microseconds.
Time measurement is not more than 1.5 sec. Warm-up time of 1 min.
The limit of additional error appliance Bridge AC R5016 caused a deviation from the ambient temperature (20 ± 1) ° C to any temperature within the operating temperature for each 10 K temperature deviation is equal to the value of 0,5 δ 0 for δ 0 ≥ 0,1 %, δ 0 - at 0,05% ≤ δ 0 <0.1%, and 0 at 2δ 0,02% ≤ δ 0 <0.05%.
The limit of additional error under the influence of an external magnetic field, 50 Hz, 80 A / m, as well as the bridge supply voltage deviates by ± 10% of nominal value equal to the value 0,5 δ 0.
Power supply Bridge AC R5016 is powered by the AC, 50 Hz ± 1% 220V ± 10% THD up to 5%.
Power consumption is less than 200 VA.
Dimensions 490h210h495 mm, weight 35 kg.
Mean time between failure is not less than 1250 hours
Note. 1. When tgδ> 0,05 or tgφ> 0,05 C measurement error, the Land Kuvelichivaetsya to 0,1 tgδ f or 0,1 tgφ f, where f - frequency, kHz.
2. α - the reading of the bridge without a comma.
3. A dash means that the error is not standardized.
4. The errors do not include the initial parameters of the bridge.