Radiometric devices >> measuring instruments of VSWR (R2- and others)
Measuring the standing wave ratio R2-68 (R268, R 2 68, R-2-68, R-268, R 268)
The composition of R2-68 includes:
- Generator Sweep (sweep);
- LED SWR and attenuation Ya2R-67.
Technical characteristics of the instrument measuring the standing wave ratio R2-68:
Frequency - 37.5 GHz-53, 57GGts;
VSWR measurement range - 1,1-5,0;
- VSWR - ± (5K + 5)%;
- Attenuation - ± (0,05 A x 0.75) dB;
The limits of the weakening of measurement - 30dB ... 0dB;
Error of frequency P2-68 - less than ± 1%;
- Sweep - 495h475h175mm;
- Indicator - 495h475h175mm;
- Sweep - 23kg;
- Indicator - 20kg.