High-frequency signal generator G4-164 (G4164, G 4164, G-4-164, G4 164)
High-frequency signal generator G4-164 is designed to control, configure and test various radio devices operating in manual control mode and in automatic control mode via the public (CPC).
The device can be a source of unmodulated signal and unsized, used as a local oscillator at the frequency of different transformations.The generator provides a measurement of the amplitude-frequency characteristics of various devices, the actual sensitivity curve and fidelity receivers, running in the unmodulated oscillations (NC), amplitude modulation (AM), frequency modulation (FM) and Pulse Width Modulation (IM).
High-frequency signal generator G4-164 is designed to work in laboratories and workshops, test agencies, repair shops.
The device is a source of high-frequency signal in the frequency range from 0.1 MHz to 639.999 MHz, running in HK, AM, FM and IM. Installing the required operating mode by using the keyboard controls on the front panel of the generator.Enter the required values of parameters such as frequency, output voltage, the coefficient of amplitude modulation, frequency deviation shall be made either through the keypad or by means of sensor kvaziplavnoy settings on the front panel of the generator.
Established parameters of the device high-frequency signal generator G4-164 can be stored in internal memory for the number assigned to a subsequent call.You can save up to 63 sets of parameters of the generator. These transactions are running embedded micro-processor-based computer N1830VE51 that can significantly reduce energy consumption and improve reliability and manufacturability compared with the base model of the generator.
Modulation of the carrier frequency is carried out as from an internal source of modulating oscillations and from the outside. Possible modulation of two parameters simultaneously: the frequency and amplitude..
Technical characteristics of the instrument high-frequency signal generator G4-164:
Frequency range - from 0.1 MHz to 639.999 MHz;
Readability of frequency tuning device generator G4-164 High-frequency signal:
- In the range from 0.1 MHz to 160 MHz - 0.1 kHz;
- In the range from 160 MHz to 639.999 MHz - 1.0 kHz;
Acceptable basic error limit frequency setting - no more than ± 1,5 · 10 -5%;
The limits of adjustment of output voltage into 50 ohms and 75 ohms in the "NC" and "World Cup":
- In the frequency range from 0.1 MHz to 639.999 MHz with a resolution of 0.1 dB - from -149.9 dBV to 6dBV (from 0.032 × 10 -6 to 2B);
- In the frequency range up to 640MGts 960MGts in increments of 0.1 dB - from -149.9 dBV to 0dBV (from 0.032 × 10 -6 V to 1 V);
- At frequencies above 960MGts - not less than 0.5;
Limits output voltage adjustment device signal generator G4-164 high-frequency loads to 50 ohms and 75 ohms in the "AM" and "TV" - from -149.9 dBV to 0dBV (from 0,032 ∙ 10 -6 V to 1 V);
Acceptable basic error limit set the reference level output voltage 0.1 V:
- 50 Ohm - no more than ± 1,0 dB;
- At 75 Ohm load - no more than ± 1,5 dB;
The instability of the output voltage reference signal generator unit high G4-164 for any 15 minutes of work after samoprogreva within 30 minutes - no more than 0.1 dB;
The frequency of the internal modulation source in the "AM" - 50Hz, 200Hz, 300Hz, 400Hz, 1000Hz; 2500Gts; 3400Gts; 10000Gts;
Frequency of the external modulation source device signal generator G4-164 high-frequency mode in the "AM" - from 30Hz to 60000Gts;
The limits set frequency deviation as a function of carrier frequency corresponds to the values:
- Carrier frequency:
- From 639.999 MHz to 320MGts - from 0.50 kHz to 995.0 kHz;
- From 160 MHz to 319.999 MHz - from 0.20 kHz to 500.0 kHz;
- From 80 MHz to 159.999 MHz - 0.10 kHz to 250.0 kHz;
- From 40 MHz to 79.9999 MHz - 0.05 kHz to 125.0 kHz;
- From 20 MHz to 39.9999 MHz - 0.05 kHz to 50.0 kHz;
- From 19.9999 MHz to 14MGts - from 0.05 kHz to 25.0 kHz;
- From 0.1 MHz to 13.9999 MHz - 0.05 kHz to 99.5 kHz;
The margin of error set the amount of frequency deviation:
- In the range of modulating frequencies from 300Hz to 3400Gts - no more than ± 10%;
- The rest of the range of modulating frequencies - no more than ± 15%;
Harmonic envelope frequency-modulated signal devices rf signal generator G4-164:
- At modulation frequencies from 100 Hz to 20,000 Hz and the significance of the deviation Δf = 0,1 · Δfmax - less than 1%;
- At modulation frequencies from 30Hz to 60000Gts and maximum deviation, depending on the frequency - 2%;
Maximum permissible error at reference level pulse width modulation mode - no more than ± 2,5 dB;
Operation mode setup time unit signal generator high G4-164 - 30 minutes
- AC - 220V ± 22V;
- Frequency - 50Hz ± 0,5 Hz;
- AC - 220V ± 11V, 115V ± 5,75 V;
- Frequency - from 372Gts to 412Gts;
- The harmonic content - up to 5%;
Time of continuous operation of the device high-frequency signal generator G4-164 - at least 16 hours;
Power consumption - less than 87V · A;
Mean time unit signal generator G4-164 high-failures - at least 6,000 hours;
Dimensions - 486h173h482mm;
Weight - not more than 22kg.
The manufacturer guarantees the quality of the instrument high-frequency signal generator G4-164 technical requirements specifications, subject to the consumer the terms and conditions of storage, transportation, installation, operation, established specifications and operational documents.