High-frequency generator G4-142 (G4142, G4 142)
High-frequency generator G4-142 is designed for use as a source of microwave oscillations when setting up, adjusting and testing of wireless devices millimeter waves.
The models applied electrical frequency tuning. Set the frequency of the signal generator unit high-G4-142 is code-switch located on the front panel.
Instruments are made primarily on the semiconductor devices and circuits with a partial application of EEC.
Technical characteristics of the devices High-frequency generator G4-142:
Frequency - 53.57 GHz-78 33GGts;
Error of frequency - no more than 1.5%;
Frequency instability of the device high-frequency signal generator G4-142 for 15 minutes - not more e 15MHz;
Spurious frequency deviation in the CW mode - 6 MHz;
Smooth frequency detuning device high frequency signal generator G4-142 - ≥ 100 MHz;
Power output - ≥ 4 ∙ 10 -3 W;
The limits of output power control unit high-frequency signal generator G4-142 - 0dB-30dB;
Mode of internal IM:
- Repetition - 1 kHz ± 0,2 kHz;
- Duty Cycle - 2 ± 0,4;
Output VSWR - 1.5;
Power flux density of the device high-frequency signal generator G4-142 - 0,5 ∙ 10 -2 W / m 2 (at 1 meter);
Acoustic noise - 60 dB (at 1 meter);
Power supply is a high-frequency signal generator G4-142: - 220 V, 50 Hz, 4000 Hz;
Power Consumption - 300V ∙ A;
Dimensions of the generator - 490h175h475mm;
Weight of generator G4-142 - 23kg.