Voltmeter E325 (E 325, E-325, E325, E 325, E-325; Є325; Є 325; Є 325, E325, E 325, E-325)
Voltmeter E325 - panel meter electromagnetic system is designed to measure current and voltage in alternating current frequency of 50 Hz - 1000 Hz.
Accuracy class - 1.5.
Front dimensions - 120 × 120 mm.
In Voltmeter E325 used in the state metrological control and supervision.
Voltmeters E325 allow basic measurement error does not exceed ± 1,5% of full scale value.
Changing E325 readings due to abnormal ambient temperature of 20 ° C (in the range of -40 º C to +50 ° C) does not exceed ± 0,8% - for Class 1.5 for every 10 degrees.
Voltmeter E325 suitable for use at temperatures between -50 º C and +60 º C. Relative humidity should be no more than 95% (at +35 º C).
E325 devices made in the body splash proof.
Positioning and mounting E325
Ammeters and voltmeters E325 E325 designed for flush mounting on vertical or horizontal panels (panels) made as of the magnetic and non-magnetic materials.
When installing the E325 need to adhere to the following rules:
- Installation of equipment should be made carefully, without distortions;
- Fastening voltmeters E325 panel must be rigid and not create additional stress;
- VoltmetrE325 the installation should be located away from sources of strong magnetic fields with a strength greater than 0.4 kA / m;
- Appliances intended for inclusion in the network with a voltage of more than 650V, mounted on insulators of reach to touch.
The manufacturer guarantees that the device specification requirements under normal use, storage, transportation and installation.
Voltmeter E325 meets the requirements for ensuring safety of life, health, consumer and environmental protection.